A Dual-Parameter Fusion Distributed Optical Fiber Sensor System for Oil and Gas Pipeline Monitoring

For oil and gas pipeline monitoring applications, this paper proposed a dual-parameter fusion distributed fiber optic sensor system that enables distributed temperature and distributed vibration measurements in a single fiber. Through the fiber-scattering spectrum time domain detection combined with coded pulse sequence and Raman scattering spectrum is obtained, which realizes high-resolution temperature measurement and wide-band vibrational wave measurement. The experimental results show that, on 10 km optical fiber measurement, temperature resolution up to 0.1˚C and vibration response frequency range 20 Hz - 5 kHz. This sensing system achieves temperature and vibration dual-parameter measurements with fiber optics, greatly simplifying the system and facilitating installation and it can be widely used in oil and gas pipeline monitoring.


Introduction
Oil and gas pipelines can easily lead to pipeline leakage due to geological disasters, man-made damage and corrosion of pipelines themselves. At present, commonly used pipeline monitoring methods include: ultrasonic method, negative pressure wave method, pressure gradient method, and the like. However, these methods are not ideal in practical applications, can only be used for pipeline leakage alarm and cannot be used for early warning monitoring [1].
The distributed optical fiber sensor has the characteristics of long measuring How to cite this paper: Li, B.J., Wu, H.L., Xie, Y.Q. and Dai, Z.Y. (2019) A Dual-Parameter Fusion Distributed Optical Fiber Sensor System for Oil and Gas Pipe-pipeline leakage monitoring. The distributed optical fiber temperature sensors (DTS) and distributed optical fiber vibration sensors (DVS) were also been verified by preliminary engineering application in oil and gas pipeline leakage monitoring [2] [3] [4] [5]. If a leak in the oil and gas pipeline, the temperature will change at the leak. The distributed fiber temperature sensor can detect the temperature change to realize the leak alarm detection. Shown high stability and low false alarm performance, but need long response time to leakage events [6]. The distributed optical fiber vibration sensors were used to monitor the oil and gas pipelines, which has the advantages of fast response speed, third-party intrusion early warning monitoring, etc., but is easily interfered by the external environment, prone to false alarm [7]. This paper proposes a dual-parameter fusion distributed optical fiber sensing system that combines fiber-scattered spectral time domain detection with pulsed coded sequence detection to achieve high signal-to-noise ratio acquisition of Rayleigh scattering and Raman scattering simultaneously. High-precision distributed temperature measurement and high-sensitivity distributed vibration disturbance measurement was realized. The fusion of the two parameters of temperature and vibration can overcome the shortcomings of the two parameters and improve the reliability of oil and gas pipeline safety monitoring.

The Principle of Coding Time Domain Detection Distributed Optical Fiber Sensor
In order to realize wide-band detection, the distributed optical fiber vibration sensor cannot perform the averaging processing of the backscattered light. It is necessary to increase the signal-to-noise ratio of the scattered signal by increasing the peak power of the detected pulse light; however, the narrow linewidth light source used in the system is easier. Producing a non-linear effect that is unfavorable to the test requires controlling the peak power of the optical pulse to a low level. In order to solve this contradiction, pulse coding technology can increase the total energy of the injected fiber and increase the intensity of the scattered light signal by increasing the number of emitted light pulses without increasing the peak power of the optical pulse.
Up to now, the coding methods used in engineering practice mainly include Simplex code, Golay code and Complementary Correlated Prometheus Orthonormal Sequence (CCPONS), as shown in Figure 1, comparison of coding gains of three different coding modes. In this paper, we choose CCPONS, because CCPONS can improve the SNR most.
The block diagram of the dual-parameter fusion fiber optic sensing system is shown in Figure 2. The entire sensing system consists of two parts: DVS and DTS.
Among them, NLL (narrow linewidth laser), AOM (acoustic light modulator),  Here, the power of narrow linewidth laser is 10 mW, the line width is 1 kHz; the detector bandwidth is 50 MHz; the data acquisition card sampling rate is 100 MS/s, the sampling precision is 16 bit; the optical pulse generated by the photoelectric modulator has a minimum pulse width of 10 ns.
When sensing system is working, the DVS and the DTS are time-division multiplexed by the photoswitch, the sensing fiber was alternately connected to the DVS or DTS, complete distributed measurement of vibration and temperature.

CCPONS Pulse Code Distributed Optical Fiber Temperature Sensor
The distributed temperature measurement can be realized by the extraction and detection of Raman scattering light signal. The system composition is shown in the part of ROTOR (lower part) in Figure 2. The anti Stokes light is separated by optical filtering for photoelectric detection. Assuming that the peak power of the optical pulse in the injected fiber is P 0 and the spontaneous Raman scattering occurs at the distance z, the optical power of the backward Raman scattering signal obtained at the injection end of the optical pulse is: [ ] where, ν ∆ is the molecular vibration frequency in the optical fiber, h is Planck constant, K is Boltzman constant. The traditional distributed light temperature sensor uses the two-way demodulation of Stokes light and anti Stokes light.
Combined with the digital processing method, formula (1) can realize the single optical path demodulation. Single optical path demodulation can significantly eliminate the nonlinear effect and ensure long-distance and high-resolution temperature measurement.
When the high-pressure gas in the natural gas pipeline leaks, the temperature of the objects around the leakage hole will drop rapidly. By monitoring the abnormal temperature changes along the natural gas pipeline using the communication optical cable, the leakage event on the gas transmission pipeline can be detected and the location of the leakage event can be accurately located.

CCPONS Pulse Coded Distributed Fiber Optic Vibration Sensor
The structure of the pulse coded distributed fiber-optic vibration sensor is shown as DVS part in Figure 2 (upper part). When natural gas leaks, the natural gas will friction with the leakage wall, which induce stress waves on the pipe wall to make pipe vibrate. The vibration generated by the pipe will act on the fiber ( ) ( ) ( )   The experiment results showed that the DTS detected an abnormal decrease of the temperature at 7798.5 m, and the temperature dropped by 6.03˚C at most. The DVS module detected an abnormal vibration signal at 7796 m, and the vibration signal frequency is concentrated in the range of 4 kHz to 4.5 kHz. It is clear that the DTS module and the DVS module are both detect the abnormal signal generated by leakage. The leakage positions detected by the two sensing modules are consistent, and the positioning error is less than 4 m.

Conclusion
In this paper we proposed a dual-parameter distributed optical fiber sensing system that combines fiber-optic scattering spectrum time-domain detection with pulsed coded sequence detection to achieve high signal-to-noise ratio (SNR) acquisition of Rayleigh scattered light and Raman scattered light with one single fiber. The experimental results show that the measurement distance is 10 km, the temperature resolution is 0.1˚C, and the vibration response frequency range is 20 Hz -5 kHz. The system combines two parameters measurement functions onto the same fiber, simplifying the test system, facilitating the application and installation, and reducing monitoring costs. In summary, the sensor system proposed in this paper has obvious technical advantages compared with other traditional leak monitoring systems, and it has great application potential in the field of oil and gas pipeline leakage monitoring.